Technical Service Bulletin Test Template
Thank you for being a valued customer of Canon Medical. In the ever-changing world of healthcare technology and information security, Canon Medical Informatics continues to advance our products to align with industry changes and to introduce new features, provide workflow improvements, enhance clinical capabilities and improve cyber security.
Technical Service Bulletin: Duplicate Series Instance UID’s
|Date||Sept 9th, 2020|
|Audience||Internal- Vital and Canon|
|Deployment(s) Type||All domain joined deployments, especially enterprise, which are always domain joined. Domain joined extend and enterprise single server (ESS) deployments are also impacted.|
|Description||When a study is imported and contains a series whose series instance UID matches the series instance UID of a series in another study that has already been imported, Vitrea will stop processing all incoming DICOM. This can happen when the series with a duplicate series instance UID is the only series in a given association or is the last series processed in a given association.|
Push each study (and all their series) in a single association (like via MediaManager). This will work unless the duplicate instance UID series is the last one to be processed in the association. This issue is unlikely to occur since the duplicate UID is detected before additional processing is complete and this bad series will most likely be processed before all the other ones have completed in the association.
Note: If a site experiences this issue and dicom processing has stopped all hung AssocServant process must be killed and Rcv3DProcessor service must be stopped. The Dicom.Association table should be cleared and the DICOMReceive folder should be emptied. Then Rcv3Processor can be restarted and studies can be pushed again. If a study has a duplicate series instance UID of an already existing series, this situation will occur again. The workaround can be used to push studies that have a series with a duplicate series instance UID.
*New hardware and Professional Services charges may apply
Use this guide to locate your software version and have the information ready for your IT department and your Canon Medical support team.
An upgrade may be necessary for you to have the highest security and full functionality on the latest product version. This may require upgraded hardware and professional services.